Kinematic viscosity apparatus
The instrument is designed and made as per the T0619 "Asphalt Kinematic Viscosity Test (Capillary Viscometer
Methods". It is suitable to determine kinematic viscosity of asphalt at a certain temperature.
Power supply: AC 220V± 10%, 50 Hz;
Heating power: Two grades; 1000 W for auxiliary heating and 600 W for temperature controlling
Bath temperature: Ambient temperature~180.0ºC
Temperature controlling accuracy: ± 0.1ºC;
Mercury thermometer: 1) Rod type
(2) Scale division: 0.1 ºC;
(3) Measurement range: 100~150ºC or 150~200ºC;
Bath cubage: Not less than 23 L
Sample quantity: 3 piece at the same time
Stirring motor: 6 W, 1200 RPM
Ambient temperature: -10ºC~+35ºC
Relative humidity: <85%
Temperature sensor: Industrial platinum resistance, PT100
Total Power consumption: Not more than 1800 W
Capillary viscometer: 7 pieces of Cannon-Fenske reserveflow capillary viscometer tubes in a group;
Their models are 200, 300, 350, 400, 450, 500, 600 respectively (Corresponding diameters
are 1.02, 1.26, 1.48, 1.88, 2.20, 3.10 and 4.00 mm).
The users should prepare Ubbelohde type viscometer tubes by themselves.